A Brief Overview of Dental Lasers
- Introduction to Lasers
- Advantages and Limitations of Lasers
- Points to Consider for Integrating a Laser info your Dental Practice
- Purchasing Considerations for Dental Lasers
Medical surgeons in about 1975 began using a new device that complemented and, in some cases, replaced the scalpel. That instrument was a laser, and during the 1980s, a Carbon Dioxide model was a common component in the operating suite. In 1989, the first laser specifically designed for dental use became available. Today there are two dozen indications for use with various dental laser devices; and the clinical applications continue to increase, making the laser one of dentistry's most exciting advances with unique patient benefits.
A laser produces light that distinguished from ordinary light by two properties: one, it is a single color, also known as monochromaticism; two, the light waves are all coherent, which means that each wave is identical in physical size and shape. This monochromatic, coherent wave of light energy emerges from the laser device as a uniquely efficient source of energy.
Lasers are generically named for the material contained within the center of the device, called an optical cavity. The core of the cavity is comprised of chemical elements, molecules, or compounds, and is called the active medium, which can be a container of gas, a crystal, or a solid-state semiconductor. One currently available dental laser uses Carbon Dioxide as a gaseous active medium. The other devices are either solid state semiconductor wafers made with multiple layers of metals or solid rods of garnet crystal grown with various combinations of other elements. For simplicity the semiconductor lasers are called diodes, and the crystal lasers are designated with acronyms such as Nd:YAG, Er,Cr:YSGG, or Er:YAG. Of course, individual manufacturers create trademarked model names, such as Waterlase, VersaWave, or Odyssey.
The active medium's stimulation generates a specific wavelength of non-ionizing radiation. A few lasers emit visible light the caries detecting system, for example; but nearly all the surgical lasers produce invisible infrared beams.
Each wavelength has a somewhat unique effect on dental structures, due to the specific absorption of that laser energy in the tissue. Some lasers are only absorbed by blood and tissue pigments, while others are only absorbed by water as well as hard tissue, like enamel, dentin, and bone.
More specifically, the wavelengths can be categorized into three groups:
- Diode and Nd:YAG wavelengths target the pigments in soft tissue and pathogens as well as inflammatory and vascularized tissue.
- Carbon Dioxide lasers also easily interact with free water molecules in soft tissue as well as vaporizing the intracellular water of pathogens.
- Erbium lasers (Er,Cr:YSGG and Er:YAG) are sometimes called all tissue instruments because very short temporal emission mode and absorption in the water of soft and hard tissue.
Lasers produce light energy that can be absorbed by a target tissue, and this absorption process produces a thermal reaction in that tissue. Depending on the instrument's parameters and the optical properties of the tissue, the temperature will rise and various effects will occur. In general, most non-sporulating bacteria, including anaerobes, are readily deactivated at temperatures of 50 degrees C. The inflammatory soft tissue present in periodontal disease can be removed at 60 degrees C; moreover, hemostasis can also be achieved within the same heat parameters. Soft tissue excisional or incisional surgery is accomplished at 100 degrees C, where vaporization of intra- and extra cellular water causes ablation, or removal of biological tissue. Likewise, the aqueous component of tooth structure and bone also boils at this temperature; thus cavity preparation, calculus removal, and osseous contouring can proceed.
There are two basic emission modes for dental lasers' continuous wave and free-running pulsed. Continuous wave means that energy is emitted constantly for as long as the laser is activated. Carbon Dioxide and diode lasers operate in this manner. Mechanical and electrical controls can produce a gated or chopped pulsing of the continuous output, helping to minimize the latent heat produced by the laser. Free-running pulse mode is produced by a flash lamp, where true pulses, on the order of a few ten-thousandths of a second, emanate from the instrument. Nd:YAG, Er:YAG, as well as Er,Cr:YSGG devices operate as free-running pulsed lasers.
Current dental lasers employ various means to deliver the laser energy to the tissue. Some use flexible small diameter glass fibers to deliver diode and Nd:YAG which are usually used in contact with the tissue. Erbium and Carbon Dioxide devices use semi-flexible hollow wave-guides or rigid sectional articulated arms; a few Erbium lasers do have larger more rigid glass fibers. Some of these systems employ additional small quartz or sapphire tips which attach to the operating handpiece, and other systems simply are used out of contact with the tissue.